I just received an ASi 1600 and all went well with the imaging session the first night. Last night, the centering action and plate solves seem to result in a large shift in the image location. When I went to calibrate and register all of the subs, a rather large portion of the image was cropped out. There were no changes in any of the settings from the night before. I don't have access to the logs right now, just curious as to whether any one else has experienced something similar to this with the 1600. Previously, I have been using an STT 8300m with no problems.
Thanks for any help.....................Gunny