I´m installed the Seletek Armadillo Autofocus system and yesterday I tried to make de curve of autofocus in SGPRO.
Apparently the parameters are ok, I will like to put some screenshots or pass to somebody the screenshots that I made yesterday to see more clearly what’s is the problem.
I could make ones the curve nearly fine but a lot of times I doesn’t , it makes a line, or a half curve of autofocus.
I focused my hand and then after I run the Autofocus system.
My equipment is a Takahashi FSQ106 and Autofocus system SELETEK Armadillo 2.
The best way It will be to send by email the screenshots to understand better the problem.
You can upload screenshots directly in this editor (7th icon from the left hand side).
I use a Takahashi FSQ 106N with original rack & pinion focuser and a Robofocus stepper motor with no problem. What focuser are you using?
In order that Autofocus works flawlessly, it is important to set the focuser backlash in SGP, R & P focusers have always backlash.
I used Seletek from Lunatico.
Would you mind explain with more clearly "it is important to set the focuser backlash in SGP, R&P focusers?
Hi I use the seletek controller and an fsq 106ed with seleteks own motor…
You may find that the issue is one of backlash setting.
I can’t remember my settings but one check worth looking at is your motor polar setting, I use 13.8 volt bi polar…
It works well for me.
I also use the Seletek controller (Armadillo2). In my case, a stepper motor from Robofocus is used.
What I wanted to know is whether the focuser of your refractor is Takahashi’s original rack & pinion focuser or whether the focuser was exchanged (e.g. for a Crayford focuser). This has nothing to do with the stepper motor.
As I wrote, R + P focusers always have some backlash. This has to be compensated, otherwise the Auto Focus will not work properly. SGP has a nice feature called “backlash compensation” for this purpose. In SGP, select Control Panel/Focus and select the button ‘Other’. Select ‘Focuser backlash compensation’, Compensation direction ‘IN’ and input the number of steps in the input field that is labeled ‘Compensation step size’.
The number of steps must be determined before. Move the focuser OUT by say 200 steps, read the focuser position and write it down. Then move it IN by a much smaller number of steps (say 20) and simultaneously observe whether the focuser actually moved. Continue to move the focuser IN by this small number of steps until you detect a movement of the focuser. Now read the focuser position again and subtract this value from the initial reading. That difference is your focuser backlash in steps. You should add some steps as a precaution, a too large value will not do damage, however, the value inputted in SGP as backlash compensation must not be too low.
I strongly advise you not to alter the ‘Advanced parameter’ of Seletek’s Configuration of the Focuser Program. If you experiment with these settings without knowing what you are doing, you possibly will damage the stepper motor. These settings have nothing to do with backlash compensation.
Thank you very much for both replaies!
Well I have follow the steps for calculate the backlash and it takes values of 100/120 steps??? Could be this numbers correct??
Tomorrow I will try to measured with more accurecy with a comparator.
100 or 120 steps is a lot of backlash. On the other hand, if everything is looking great, only this (backlash) or a mechanical problem can be the cause.
What’s been suggested is great, configure the motor as bipolar (it will have double the torque, sometimes needed for the beefy FSQ focusers) and configure the backlash - better a slowly low value than a higher one.
Also make sure the physical couplings are not slipping.
That should do.
Well today I have made a precision measures.
First with the tube horizontal the backlash has been 100 steps.
With the tube at 43 grades the backlash has been 80 steps.
And finally with the tube at 90 grades the backlash has been 60 steps.
The motor is configured in bipolar.
and I’m sorry, I don’t get the email notifications so never find out when messages here are updated
Checking my older message - now I disagree with myself. I mean, 100 or so is a lot for the motor (motor+gearing) backlash; including the one from the r&p focuser, may be perfectly normal.
My apologies for the confusion.
Thanks for your answer!!
Sorry for the delay. I’m really getting crazy with the autofocus. I have a Takahasi FSQ 106 flourite tube. On saturday night I made an easy exercise, I focused with Bathinov mask and after I run the autofocus and put the mask to check if it the autofocus was working properly, the result was that it wasn’t in the proper focus point.
I couln’t made more.I’m really worried about this. I put the telescope working indoor to check if the movement in-out is fine and I saw is fine and the distance to inside and outside is the same.
Not sure if the parameters that I have in the Sgp and Seletek are ok.